could anyone help me to clarify these subjects ?
Does İvent has single hose system or dual hose system ? İvent has new vent mode adaptive bilevel ventilation.is this mode similar to BiPAP(Respironic) or BIPAP(as Drager have). is İvent suitable to use for infant patients ? what is the minumum tidal Volume we can set se up ? also can we use İvent for CPAP applications ? what is the max PEEP we can set ?
thank you very much for your response in advance.
The iVent Adaptive BiLevel mode is similar to Respironics. The Ventilator does not have an active exhalation valve. The unit has internal PEEP up to 20cmH20 and can be used in the CPAP/PSV mode. Currently, the smallest Vt in volume modes is 100ml.